A Failure Study of a Locking Compression Plate Implant
In this case study a failed locking compression plate was investigated. Such plating systems are used to provide the stability to fractured bone and fixation. The locking compression plate had been separated in two pieces. One of the fracture surfaces from the failed component was investigated for surface topographical features. The visual, optical and scanning electron microscopy results indicated the presence of beach marks, intermetallic inclusions, corrosion pits and striations indicating fatigue crack propagation and overload failure. Some corrosion damage also was documented on the fractography. This case study shows that corrosion may have initiated fatigue crack which grew by the activities of daily living causing the failure.
Prayson, M. J.,
& Goswami, T.
(2015). A Failure Study of a Locking Compression Plate Implant. Case Studies in Engineering Failure Analysis, 3, 68-72.